The Sine-wave Vibro Viscometer measures viscosity by detecting the driving electric current necessary to resonate two sensor plates at a constant frequency of 30Hz, and amplitude of less than 1mm.
Using a newly patented Tuning-fork Vibration Method, The SV Viscometer allows for high accuracy in a wide variety of measurement ranges, without replacing the sensor plates!
High Measurement Accuracy
Wide Measurement Range
Non-Newtonian Sample Viscosity Measurement
Foam Sample Measurement
Flowing Sample Measurement
Vacuum Fluorescent Displays
Sol and Gel Measurement
Standard RS-232C Interface
Small Sample Size
Data Collection and graphing Software
SINE-WAVE VIBRO VISCOMETER
1. RsVisco creates the real-time graph of data received from
SV-series viscometer via RS-232C.
RxVisco lets you graphically monitor in real time:
■Progress of change in viscosity during measurement
■Temperature and viscosity, graphed together
TUNING FORK METHOD (HOW IT WORKS)
Two spring plates are vibrated with a uniform frequency, the amplitude varies in response to the amount of frictional force produced by the viscidity between the sensor plates and the sample. To produce uniform amplitude, the vibro viscometer controls the electrical current that drives the vibration of the spring plates.
Because the frictional force of viscidity is directly proportional to the viscosity, the driving electric current for vibrating the spring plates with a constant frequency to produce uniform amplitude is also directly proportional to the viscosity of each sample.
The vibro viscometer measures the driving electric current, and then the viscosity is given by the positive correlation between the driving electric current and the viscosity.
SV SERIES VISCOMETERS 640SV10 SV100
SAMPLE VISCOSITY CHART centipoise
WATER 70 F 1~5
BLOOD / KEROSENE 10
ANTI-FREEZE / ETHYLENE GLYCOL 15
MOTOR OIL SAE 10 / CORN OIL 50~100
MOTOR OIL SAE 30 / MAPLE SYRUP 150~200
MOTOR OIL SAE 40 / CASTOR OIL 250~500
MOTOR OIL SAE 60 / GLYCERIN 1,000~2,000
KARO CORN SYRUP / HONEY 2,000~3,000
BLACKSTRAP MOLASSES 5,000~10,000
HERSHEY CHOCOLATE SYRUP 10,000~25,000
HEINZ KETCHUP / FRENCH'S MUSTARD 50,000~70,000
TOMATO PASTE / PEANUT BUTTER 150,000~250,000
AREA FOR NOTES
MINIMUM MAXIMUM VALUES
You can choose from three types of graphs:
■Viscosity (Y axis) - Time (X axis)
■Viscosity/Temperature (Y axis) - Time (X axis)
■Viscosity (Y axis) - Temperature (X axis)
Windows CT VISCOSITY SOFTWARE
Using an RS232C interface, the WinCT-Viscometer Software, allows the Sine-wave Vibro Viscometer to display measurement progress in real time on a PC, and easily transmit the measured results to save or analyze. Changes of viscosity and temperature over time, as well as the correlation between viscosity and temperature, can be observed visually.
2. Graphs can be overlaid in repeating measurements, in 10 colors.
3. Measured data can be saved in a CSV format file.
4. Displayed graphs can be printed with a printer via a PC.